Carbon Nanotube Forest Reflects Almost No Light
Harnessing darkness for practical use, researchers at the National Institute of Standards and Technology (NIST) have developed a laser power detector coated with the world's darkest material—a forest of carbon nanotubes that reflects almost no light across the visible and part of the infrared spectrum.
NIST will use the new ultra-dark detector, described in a new paper in Nano Letters,* to make precision laser power measurements for advanced technologies such as optical communications, laser-based manufacturing, solar energy conversion, and industrial and satellite-borne sensors.
Inspired by a 2008 paper by Rensselaer Polytechnic Institute (RPI) on "the darkest man-made material ever,"** the NIST team used a sparse array of fine nanotubes as a coating for a thermal detector, a device used to measure laser power. A co-author at Stony Brook University in New York grew the nanotube coating. The coating absorbs laser light and converts it to heat, which is registered in pyroelectric material (lithium tantalate in this case). The rise in temperature generates a current, which is measured to determine the power of the laser. The blacker the coating, the more efficiently it absorbs light instead of reflecting it, and the more accurate the measurements.
The new NIST detector uniformly reflects less than 0.1 percent of light at wavelengths from deep violet at 400 nanometers (nm) to near infrared at 4 micrometers (μm) and less than 1 percent of light in the infrared spectrum from 4 to 14 μm. The results are similar to those reported for the RPI material and in a 2009 paper by a Japanese group. The NIST work is unique in that the nanotubes were grown on pyroelectric material, whereas the other groups grew them on silicon. NIST researchers plan to extend the calibrated operating range of their device to 50 or even 100 micrometer wavelengths, to perhaps provide a standard for terahertz radiation power.
NIST previously made detector coatings from a variety of materials, including flat nanotube mats. The new coating is a vertical forest of multiwalled nanotubes, each less than 10 nanometers in diameter and about 160 micrometers long. The deep hollows may help trap light, and the random pattern diffuses any reflected light in various directions. Measuring how much light was reflected across a broad spectrum was technically demanding; the NIST team spent hundreds of hours using five different methods to measure the vanishingly low reflectance with adequate precision. Three of the five methods involved comparisons of the nanotube-coated detector to a calibrated standard.
Carbon nanotubes offer ideal properties for thermal detector coatings, in part because they are efficient heat conductors. Nickel phosphorous, for example, reflects less light at some wavelengths, but does not conduct heat as well. The new carbon nanotube materials also are darker than NIST's various Standard Reference Materials for black color developed years ago to calibrate instruments.
Source: NIST /...
The Institute of Nanotechnology puts significant effort into ensuring that the information provided on its news pages is accurate and up-to-date. However, we cannot guarantee absolute accuracy. Consequently, the Institute of Nanotechnology disclaims any and all responsibility for inaccuracy, omission or any kind of deficiency in relation to the news items and articles hosted herein.
- 16 October 2014Glasgow nanofabrication expert receives prestigious award
- 26 September 2014On the Road to Artificial Photosynthesis
- 23 September 2014A nanosized hydrogen generator
- 03 September 2014New Synthesis Method May Shape Future of Nanostructures, Clean Energy
- 14 August 2014“Trojan horse” treatment could beat brain tumours
- View All